Olanzapine/Desmethylolanzapine Plasma Calibration Standard
Olanzapine/Desmethylolanzapine Plasma Calibration Standard, lyophilised
This lyophilised calibrator from Chromsystems is based on human plasma and should be used according to the instruction manual. It is designed to calibrate the measurement method in clinical diagnostics for the quantitative determination of olanzapine and desmethylolanzapine in serum/plasma. While following the instructions the calibrator should be handled and measured in the same manner as a patient specimen.
|Content||5 x 2.5 ml (lyoph.)|
Please note: Concentration values shown here are typical. For lot specific concentration values please refer to the product insert supplied with the materials. The product inserts can also be requested from us directly or downloaded from our website after login.
Carefully remove the metal seal and rubber stopper from the vials. Add exactly 2.5 ml of distilled water into each vial. Return the stopper and incubate for 10 to 15 minutes at room temperature. Swirl the vials to dissolve the contents until homogeneity.
Stored below –18 °C and unopened, the lyophilised calibrator is stable until the expiry date specified on the vial labels. The reconstituted calibrator can be stored up to 1 week light protected and tightly capped at +2 to +8 C. If the calibrator cannot be used within this period, it should be aliquoted and stored frozen below -18 °C (up to 3 months). For more details about storage conditions please check your instruction manual.
|Method of Analysis||HPLC|
|Parameter||N-Desmethylolanzapine, Olanzapine, Quetiapine|